The more monitors we test, the more we can see that off-axis viewing performance is dependent not only on pixel structure (IPS, PLS, TN, etc.) but the backlight technology as well. And we can see that the anti-glare layer makes a difference too.
Remember back to ViewSonic's VP2772? If you think the PA272W’s off-axis pictures look a lot like that screen, we won't be surprised. After all, they both employ the same LG panel.
You can see a slight red shift in the top and bottom shots, and almost no tint in the side-to-side views. Moreover, there’s little loss of detail. Of course, for critical work where you plan on sitting front and center anyway, consider a hood for the best possible image quality.
Screen Uniformity: Luminance
To measure screen uniformity, zero and 100-percent full-field patterns are used, and nine points are sampled. First, we establish a baseline measurement at the center of each screen. Then the surrounding eight points are measured. Their values get expressed as a percentage of the baseline, either above or below. This number gets averaged. It is important to remember that we only test the review sample each vendor sends us. Other examples of the same monitor can measure differently.
Our results reflect both the On and Off settings of the uniformity compensation. There are five levels ranging from strong (5) to weak (1). We used 5 and Off in our tests. You’ve already seen how level 5 affects contrast. Now it’s time to see if that reduction is worth the improved screen uniformity.
The black field measurement with compensation off is about average among the monitors we’ve tested. We couldn’t see any hotspots on our press sample, but the C6 meter tells us that the upper-right and lower-left corners are a tiny bit brighter. Turning the compensation on yields a two-percent improvement.
Here’s the white field measurement:
Screen uniformity is just shy of seven percent in the full-white field pattern. Again, you wouldn't be able to see that with a naked eye. Our meter says the center portion of the screen is brightest and the upper-left zone is a bit dimmer. I stand by my preference of leaving the compensation off, and enjoying better contrast and black levels.
Screen Uniformity: Color
To measure color uniformity, we display an 80-percent white field and measure the Delta E error of the same nine points. Then we subtract the lowest value from the highest, giving us a result. A smaller number means a display is more uniform. Any value below three translates to a variation that is invisible to the naked eye.
The compensation feature helps a little with color uniformity as well. Both results are well below the threshold of visibility, however. Regardless of signal level, the PA272W renders a perfectly smooth-toned image over the entire screen. The performance is excellent.
- NEC PA272W 27” QHD Professional Monitor Review
- Packaging, Physical Layout, And Accessories
- OSD Setup And Calibration Of The NEC PA272W
- Calibration With SpectraView
- Measurement And Calibration Methodology: How We Test
- Results: Brightness And Contrast
- Results: Grayscale Tracking And Gamma Response
- Results: Color Gamut And Performance
- Results: Viewing Angles And Uniformity
- Results: Pixel Response And Input Lag
- NEC PA272W: Unparalleled Accuracy And Flexibility