Why you can trust Tom's Hardware Our expert reviewers spend hours testing and comparing products and services so you can choose the best for you. Find out more about how we test.
Transient Response Tests
Advanced Transient Response Tests
For details on our transient response testing, please click here.
Ιn these tests, we monitor the TPG-0850D-R's response in two different scenarios. First, a transient load (10A at +12V, 5A at 5V, 5A at 3.3V, and 0.5A at 5VSB) is applied for 200ms while the PSU works at 20 percent load. In the second scenario, Thermaltake's contender is hit by the same transient load while operating at 50 percent load. In both tests, we use our oscilloscope to measure the voltage drops caused by the transient load. The voltages should remain within the ATX specification's regulation limits.
These tests are crucial because they simulate the transient loads a PSU is likely to handle (such as booting a RAID array or an instant 100 percent load of CPU/GPUs). We call these tests "Advanced Transient Response Tests," and they are designed to be very tough to master, especially for a PSU with a capacity of less than 500W.
Advanced Transient Response at 20 Percent
Advanced Transient Response at 50 Percent
The +12V rail approaches 1%. This is an acceptable deviation, though in this category we'd like to see close to 0.5% readings on this rail. The 5V and 3.3V rails encounter controlled voltage drops, however the latter rail drops below 3.2V during the second test because of its low initial voltage. Finally, the 5VSB rail performs poorly in these tests.
Here are the oscilloscope screenshots we took during Advanced Transient Response Testing:
Transient Response At 20 Percent Load
Transient Response At 50 Percent Load
Turn-On Transient Tests
In the next set of tests, we measured the response of the PSU in simpler transient load scenarios—during its power-on phase.
For the first measurement, we turned off the TPG-0850D-R, dialed in the maximum current its 5VSB rail could output, and switched the PSU back on. In the second test, we dialed the maximum load the +12V rail could handle and started the 850W supply while it was in standby mode. In the last test, while the PSU was completely switched off, we dialed the maximum load the +12V rail could handle before switching it back on from the loader and restoring power. The ATX specification states that recorded spikes on all rails should not exceed 10 percent of their nominal values (+10 percent for 12V is 13.2V, and 5.5V for 5V).
The 5VSB rail behaves well. During the second test, the +12V rail's slope does a small wave that we aren't worried about. The last test is really bad though, with the +12V rail dropping its output close to zero for for several milliseconds. This is something that we didn't expect to see from a PSU that costs $180 (at the time of the review). Needless to say, such a bad result will cost the PSU several performance points. Thankfully, going from off to full load is almost impossible to reproduce under real-world conditions. But the TPG-0850D-R should still be able to handle this scenario without a problem.
Current page: Transient Response TestsPrev Page Cross-Load Tests And Infrared Images Next Page Ripple Measurements
Stay on the Cutting Edge
Join the experts who read Tom's Hardware for the inside track on enthusiast PC tech news — and have for over 25 years. We'll send breaking news and in-depth reviews of CPUs, GPUs, AI, maker hardware and more straight to your inbox.
Aris Mpitziopoulos is a Contributing Editor at Tom's Hardware US, covering PSUs.
-It's not exactly a PSU-related question but more likely a general one.Reply
From the pictures from the monitoring software, i can notice that this software is able to measure CPU and GPU temperature. Do you have any idea if such monitoring is (relatively) accurate?
-Excellent review Aris !!
in corsair link software atleast both cpu and gpu temperatures are accurate.Reply
it takes readings from the system's (actually the individual components') sensors. If those are accurate then the software's readings will be accurate as well. Normally they are accurate.Reply
That 12V transient response was certainly interesting, I don't think I've seen one do anything like that beforeReply
18886218 said:That 12V transient response was certainly interesting, I don't think I've seen one do anything like that before
18886236 said:18886218 said:That 12V transient response was certainly interesting, I don't think I've seen one do anything like that before
Go and plummet back down to almost 0V during the test
18887052 said:18886236 said:18886218 said:That 12V transient response was certainly interesting, I don't think I've seen one do anything like that before
Go and plummet back down to almost 0V during the test
I don't think you're reading the transient response graph properly. It's not a measure of voltage but a measure of ripple (well, technically that is a measure of voltage but it's more a measure of instantaneous offsets of an average, or a highly zoomed in sector).
That or I am just missing something. I probably am. Not sure what I'm missing though. Forgive my ignorance.
For this graph http://media.bestofmicro.com/ext/aHR0cDovL21lZGlhLmJlc3RvZm1pY3JvLmNvbS9TLzYvNjIxNzk4L29yaWdpbmFsLzEydl9vZmYuanBn/r_600x450.jpg I hope that I'm not reading it that badly wrongReply
yes this is something really weird indeed.Reply
Fantastic review Aris! I absolutely love reading in depth articles about anything. Terrible PSU for the price, but it's an awesome review. Congrats!Reply